Kiilto Pro Easydes Wipe
Ready to use wipes for cleaning and disinfection of non-invasive medical devices and high-touch surfaces. Quick and efficient against microbes.
- CE marked medical device, class IIa, NB 0537
Small package contains 16 wipes, large 30. Low-linting wipes are impregnated with an ethanol based disinfectant with cleaning properties. The disinfectant is quick-acting and tolerates organic stress well, efficiently destroys germs, viruses and fungi. Disinfecting efficacy is also achieved on visibly contaminated surfaces.
The microbicidal effectiveness of the product has been verified with EN- standard tests: EN 16615 with the 30 pcs wipe (16 pcs wipe pass EN16615 with S. aureus, P. aeruginosa, C. albicans), as well as tests for the liquid: EN 13727 (bacteria), EN 13624 (yeasts), EN 14348 (tuberculosis and mycobacteria), EN 14476 (adeno- and noroviruses), EN 14561 (bacteria), EN 14562 (fungi, yeasts), EN 14563 (mycobacteria and tuberculosis).
In infection risk situations, when transmission routes of microbes must be terminated on high-touch surfaces. Compatible with commonly used materials, most plastics included.
For examination equipment, medical devices and surface disinfection within health care. For example stethoscopes, handles, keyboards, switches and high touch surfaces. Excellent also within homecare.
Medical device. For professional use.
Product pH: 10
|Package Size||Product Code||GTIN|
|16 x 16 pcs||8285||6417964082852|
|18 x 30 pcs||8286||6417964082869|
At room temperature.
8285: Shelf life 1,5 years from manufacturing.
8286: Shelf life 2 years from manufacturing.
Wipe the device/surface to be disinfected thoroughly and carefully. Make sure the surfaces are well moistened with the disinfecting agent. Wait until alcohol has evaporated from the surface before the next use of the device.
Most often pre-washing or rinsing is not necessary. If the item needs to be free from all residue, rinse or wipe it with clean water after disinfection. When the disinfectant has evaporated and the surface has dried, disinfection has taken place.
Serious incidents in relation to Kiilto Pro Easydes disinfection wipes are to be reported to firstname.lastname@example.org and the local medical device authority (Fimea in Finland).
BPR: PT 2 and PT 4
Use biocides safely. Always read the label and product information before use.
Safety instructions of the product safety data sheet.
Mark the date of first use on the package with a ballpoint pen. Close the package carefully after use to keep the alcohol from evaporating.
8285: Opened package must be used within 7 days.
8286: Once opened, use within 3 months.
Follow cleaning and disinfection instructions given by manufacturer of the device to be reprocessed.
In case of Clostridium difficile epidemies use Kiilto Pro Klorilli or Kiilto Pro Erisan Oxy+ disinfectant is recommended, as spores can not be inactivated with ethanol based products.
|Ethanol (60 p-%)||Biocide|
|Tertiary butanol (< 2 %)||Denaturation agent|
|Quaternary ammonium chloride (< 0,5 %)||Biocide, cleaning properties|
|Alkylamine (< 0,5 %)||Biocide|
Wipe material polyester 60%, viscose 40 %.
8285: Wipe size 200x135 mm (16 pcs).
8286: Wipe size 285 x 200 mm (30 pcs).